2 Radiological Department, Oslo University Hospital, Rikshospitalet, N-0027 Oslo, Norway
In a prospective manner to evaluate the range of acetabular component anteversion actually achieved by the use of a cup positioner in cementless revision and primary THA.
We operated 71 patients with cementless primary THA, and 26 patients with cementless acetabular revision surgery. We aimed to obtain cup anteversion of 10 to 30° with an impactor-positioner. In all cases we used elevated liners and a ceramic head with diameter 28. At 3 months postoperatively the component versions were measured using CT with the patient in supine position.
The acetabular component version in the primary hips ranged from 28° of retroversion to 42° of anteversion with a mean of 17.4 ± 14.0°, while the cup version in the revision hips ranged from 4° of retroversion to 32° of anteversion with a mean of 15.0 ± 9.6°(p=0.427). The anteversion of 40 (56%) of the primary acetabular components were within the target zone of 10 to 30°, while 19 (27%) were below the target zone and and 12 (17%) were above the target range. The anteversion of 19 (73%) of the revision acetabular components were within the target zone, while 6 (23%) were below the target zone and 1 (4%) were above the target range. The differences in distribution between the primary and revision operations were not significant (p=0.183).
The intraoperative estimation of acetabular anteversion by free hand technique in many cases was not within the intended range of 10 to 30° in either primary or revision THA and with no differences between the two series
open-access license: This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.5/) which permits unrestrictive use, distribution, and reproduction in any medium, provided the original work is properly cited.
* Address correspondence to this author at the Orthopaedic Departement, Oslo University Hospital, Rikshospitalet, N-0027 Oslo, Norway; Tel: +4723076010; Fax: +47 23076013; E-mail: email@example.com